Datanetics DC-60 series
|This article requires additional photographic illustration — need cream, latching, brown etc|
|Sense method||Metal leaf|
|Switch mount||Plate mount|
Datanetics DC-60 series is the lower-profile, lower-cost alternative to Datanetics DC-50 series. DC-60 switches are fairly uncommon, and are best known for being used in Fluke Y1700 keyboards. In addition, they have also been used on Fluke test equipment from the era such as the high end Fluke 5440B DC calibrator. All known examples are unbranded. Unlike DC-50, this series is not known to be patented.
The slider colour was not specified, but it appears to be an indicator of the switch profile, with evidence suggesting low profile is brown and grey, lower profile is black, and lowest profile being white, with cream appearing to be a compromise between lower and lowest profiles not shown in the extant brochures. This remains pure speculation owing to no measurements yet existing for any encountered variants.
- Brown slider, low profile
- Grey slider, low profile (below)
- Black slider, lower profile (below)
- White slider, lowest profile (below)
- Black slider, latching (appears to be lower profile)
- Cream slider — this might be a fraction of a millimetre lower in profile than "low" profile, but moreover, this variant has a much narrower cross which appears to be Cherry MX mount
In 1975, only the standard (AKA lower) and low (AKA lowest) profiles existed. By 1981, a marginally taller ("low") profile had been introduced, along with a latching variant.
Officially, all switches are tactile, but the lowest-profile (snap-on) switch provided by Meryl Miller is inexplicably linear. His medium profile tactile switch is highly tactile, and late-issue ITT-made lowest-profile switches have been obtained with only weak tactility. It is not known if linear switches went into production; Sandy reports the Fluke Y1700's switches as linear with dust-impaired feel, which does not guarantee that his medium-profile switches do not share the same weak tactility as the contemporary lowest-profile switches.
Known part numbers
|Part no.||Action||Profile||Guide posts||Defined in|
|DC-61-01||Momentary||Lowest||Present||Datanetics 1975, ITT Datanetics 1981|
|DC-61-02||Momentary||Lowest||Absent||Datanetics 1975, ITT Datanetics 1981|
|DC-61-03||Momentary||Lower||Present||Datanetics 1975, ITT Datanetics 1981|
|DC-61-04||Momentary||Lower||Absent||Datanetics 1975, ITT Datanetics 1981|
|DC-61-05||Momentary||Low||Present||ITT Datanetics 1981|
|DC-61-06||Momentary||Low||Absent||ITT Datanetics 1981|
|DC-62-03||Latching||Lower||Present||ITT Datanetics 1981|
|DC-62-04||Latching||Lower||Absent||ITT Datanetics 1981|
|DC-62-05||Latching||Low||Present||ITT Datanetics 1981|
|DC-62-06||Latching||Low||Absent||ITT Datanetics 1981|
The design changed with time. The heat-stakable guide posts (fixing pins) appear to have been shortened significantly at some stage, from 2.92 mm in 1975, to 1.02 mm in 1981. The 1981 brochure still depicts long guide posts and still describes them as heat-stakable, but the specifications, which match the form most commonly found, is for posts that may only be long enough to serve as fixing pins. The diagonal corners were also removed at some point, being absent from both Sandy's Fluke Y1700 examples, and from terrycherry's cream examples.
- HP 9825A — brown gen. 2, 1979
- Fluke 1720A Programmer Keyboard — black gen. 2, 1980
- Fluke Y1700 Keyboard — black gen. 3, 1990
- Fluke 5440B DC Calibrator
Black lower profile
- Deskthority — USw LXBB03?
- Deskthority — ID this keyswitch
- AliExpress — 5pcs/lot infrequent Keyboard switch yellow shaft …
- Sandy55 — FLUKE Y1700
- Deskthority — HP 9825 keyboard restoration Posted 2014-05-05. Retrieved 2015-08-03.
- Deskthority — Fluke 1720A Programmer Keyboard - Switches?
- Flickr — Fluke Y1700 Keyboard
- EEVBlog - Repairing a 5440B/AF DC Calibrator